Please read description carefully.
Do not base your order on the picture presented.
The Welch Allyn Miller Laryngoscope Blade, Size 3 is the smart choice.
In 1990, Welch Allyn modified the Miller blade to improve its functionality and field of view. First, the base or tongue of the blade was flattened so that the tongue could be "swept" aside, rather than compressed. This was implemented due to the present technique of "sweeping" the tongue rather than compressing. The flat base helps to hold the tongue out of the field of view. Overall field of view was increased by modifying the cross-sectional area of the blade without altering width or height. This resulted in a larger field of view.
Inside Vertical: 11 mm
Inside Base to Tip: 172 mm
Other Information: The light output of the fiber optic bundles may degrade after about 1, 000 sterilizations. Fiber optic bundles may be easily removed and returned to Welch Allyn for repolishing to restore light output.
Warranty: Welch Allyn will repair or replace, free of charge, any parts of its own manufacture proven to be defective through causes other than misuse, neglect, damage in shipment, or normal wear.
Eligible for FREE or discounted ground shipping.
Medex SKU: WA-68063