Please read description carefully.
Do not base your order on the picture presented.
Use a wet tissue with alcohol to clean the probe cable and the surface of the probe body.
Wipe only with disinfectants and make sure that liquid does not enter the probe body.
For periodical cleaning of the probe body, contact your authorized service department.
The probe body contains sensitive components. Never clean the sound apertures in the probe body mechanically or with liquids.
Doing so may cause damage to the probe.
Even the slightest amount of moisture may dissolve any residual cerumen and contaminate the sensitive parts in the body of the probe.
Probe tip cleaning:
Probe tips are disposables and should not be cleaned.
Probe tips should be disposed of according to local regulations.
You should always comply with local hygienic standards.
The probe tip usually does not come into contact with the skin or secretion from the ear canal, as it is covered by the ear tip, but you should check the channels in the probe tip every time you have used the probe.
Even small amounts of cerumen or vernix can block the probe channels or be deposited on the probe tip.
Remove the probe tip from the probe body. Use the cleaning wire to clean the sound channels in the probe tip.
Always clean from the rear end of the probe tip.
Make sure you wipe any debris from the cleaning wire with a disinfectant before pulling it back through the sound channels.
The probe tip must be completely dry before placed on the probe body
Medex SKU: GNO-8-69-41700